To avoid electric shock, do not test any diode that has voltage on it. Put the fuse door back in place.
In addition to the normal safety precautions used in working with electrical circuits. If both readings are very small or 0, the device is shorted. To avoid electric shock, crraftsman not operate your meter until the battery door is in place and fastened securely. To avoid electric shock, disconnect the test leads from any source of voltage before removing the fuse door.
Touch the test probe tips to the circuit under test. Touch the black test probe tip to the negative side of the circuit.
Do not make high current measurements on the 10A scale for longer than 15 seconds followed by a 15 minute cool down period. ALWAYS discharge filter capacitors in power supplies and disconnect the power when making resistance or diode tests.
To avoid electric shock, disconnect power to the unit under test and discharge all capacitors before taking any resistance measurements. By pressing ‘print’ button you will print only current page. To avoid electric shock, disconnect the test leads from any source of voltage before removing the battery door. Page of 19 Go.
82434 some low AC and DC voltage ranges, with the test leads not connected to a device, the display may show a random, changing reading. A reference voltage, current, etc. For current measurements up to 10A DC, set the function switch to the A position and insert the red test lead banana plug into the 10A jack.
Your browser security needs to be updated
Large voltage surges may occur that can damage the meter. Read the frequency or duty cycle in the display. Do not make current measurements on the 10A scale for longer than 30 seconds.
Got it, continue to print. As a result, the reading may show 0 volts when the outlet actually has voltage on xraftsman. The batteries should be 823344. Put the battery door back in place. Temperature extremes can shorten the life of the electronic parts and distort or melt plastic parts. Check to be sure the batteries are properly installed. Remove old or weak batteries so they do not leak and damage the unit. Secure with the two screws.
NEVER connect the meter leads across a voltage source while the function switch is in the current, resistance, or diode mode.
If your meter does not work properly, check the fuses and batteries to make sure that they are still good and 82344 they are properly inserted. For measurement situations requiring that a range be manually selected, perform the following: Call our Customer Service Line Apply power to the circuit.
This symbol adjacent to one or more terminals identifies them as being associated with ranges that may, in normal use, be subjected to particularly hazardous voltages. Comments to this Manuals Your Name.
Be sure to observe the correct polarity red lead to positive, black lead to negative.
CRAFTSMAN Calibration – Micro Precision
Insert the black test lead banana plug into the negative COM jack and the red test lead banana plug into the positive V jack. To avoid electric shock, disconnect power to the unit under test and discharge all capacitors before taking any capacitance measurements.
This warranty gives you specific legal rights, and you may also have other rights which vary from state to state.