Accutrend GC Blood Pressure Monitor pdf manual download. Since each code strip and test Glucose ® Accutrend, GC, GCT, DM CODE 50 TESTS. Description Reviews (0). ACCUTREND GCT PLUS BLOOD ANALYSIS METER. Test Report For: Glucose User Manual. Optional Request: In Exchange For. DownloadAccutrend gct user manual. Semoga Download driver keychain coby dp usb win xp, Hi osvaldorodriguez, are you If you are still a Windows user.
The time must also be changed from summer to win- ter time and vice versa, and 29 February must be entered in a leap year. For further information on this Weight: View digital manual for this product.
Speedy as when when you delivered to me in Kaduna. Page 8 If at any acchtrend during the function test the display does not correspond to either of the diagrams on the left, please notify the Roche Diagnostics service see pages 97 and To print the manual completely, please, download it. If the strip code has not been read in correctly, error messages appear e. Select the relevant directory by pressing for glucose values or for cholesterol values. System Components For further information please ask your For disposal of the meter the rele- pharmacist or your Roche Diagnostics office.
Press the button MEM and hold it down to run through all memory positions. Measure three important health parameters – cholesterol, blood glucose and triglycerides in one easy process, leaving you in better control of your health. The Accuchek GCT System lets you aaccutrend track of the trend by using its memory function to store your measurements, up to 50 glucose, 20 cholesterol and 50 triglyceride values.
Don’t have an account? Plausibility check 37, 69 Practising measurement 29ff, 59 Retrieval of results from xccutrend 49ff. The abbreviations used by the Accutrend. This is one major self-testing device to put you in good zccutrend of the direction of your health, even as you strive for the accomplishments you desire. The test strip is not suitable for the meter.
ROCHE ACCUTREND GC USER MANUAL Pdf Download.
Keep it up and make Nigerians proud. During the last four seconds a series of beeps is yct one beep per second followed by a longer beep at the end of the measurement.
Don’t show me this message again. Prior to that, I wasn’t sure you could order goods online on any site in Nigeria and get delivery.
CODE The display now counts down in seconds to. Contact the Roche Diagnostics service centre. It should also be cleaned: If other messages appear on the display, consult Sec- Too little blood applied — The used test strips and consumables e. This helps the doctor when results are reviewed The GCT System, while greatly suited for clinical testing, is also personal testing made easy.
Safety warnings such as are also stored. Troubleshooting Move the meter to a dry environment and wait a while before switching it on again.
October Softclix are trademarks of a Member of the Roche Group. Logging event code facility, too: Page 46 Move the meter to a different place see Section 5. If the instrument has not been coded before, the display shows.
Document the circumstances of your measurement example: And if you accutend running a screening campaign, or just for use in your clinic, this is a tool for quick, reliable testing.
Illustrations of the meter are shown on the fold-out page. Table of Contents Add to my manuals Add.
Humidity In the presence of high humidity or large temperature changes condensation may form on the meter, e. If the result is again outside the specified range, consult your nearest Ch. After 2 minutes with test strip inserted; after 1 minute if no button was pressed and no test strip has been inserted.
Get the results in a matter of seconds and minutes: When all positions have been shown the display returns to To count down from any value stored: If too little blood is applied, do not try to spread it or apply a second drop as this may also produce incorrect results.
A performance check should be carried out: The minutes flash on the display.
Roche Accutrend GC User Manual
You have the following possibilities: Blood collectionTiong Bahru Road They are essential to ensure precision and reliability of the measurements. On target high relevance: Manufacturer’s Warranty And Guarantee Press the SET button briefly. Move the meter to a different place see Section 5. If any of these items are missing or damaged please contact your supplier.
Liquid crystal display Automatic cut-off: The test strip is too old or has been incorrectly stored.